The botiss grafter is an applicator specifically designed for dental use. It enables convenient hydration of particulate bone substitute materials or mixtures thereof and facilitates their efficient delivery to the defect site. Hydration is easily performed with the help of the integrated sieve cap.
Matching the treatment concept
Aligned with the treatment concept and the individual needs of surgeon and patient, the botiss grafter enables application with the selected bone substitute material. For comfortable filling, the botiss grafter is equipped with an integrated funnel.
The indicated maximum filling height should not be exceeded so that there is sufficient volume for the plunger rod to move upwards and downwards for optimal hydration after filling.
The maximum filling height is marked on the syringe body with an arrow and the word “MAX”.
For hydration, at least 3 up- and downward movements of the plunger rod with visible swirling of the bone substitute material have to be performed. Subsequently, excess liquid has to be removed by pushing down the plunger rod without compressing the bone substitute material excessively.
Removal of the sieve cap
After hydration, the sieve cap is removed by snapping it off (e.g. by using the thumb).
The botiss grafter is provided sterile, it is intended for single use only. It must not be resterilized or reused.
With our international network of distribution partners, we are near you in over 100 countries worldwide. In addition to our 360° productportfolio, we offer service, scientific advice and exchange, training and events directly on site from a single source.